Total posts : 45366
“Farther down in that thread another person posted his results.”
The second set of results are more accurate than the first. That is comparing to the tests I ran going straight into the Motorola R 2012 D analyzer through a 2 foot long piece of RG-214 into the power RF input terminating with the analyzer’s internal 50 ohm dummy load, which was checked and measured precisely 49.97 ohms.
Care to see the pictures? Well maybe not, photographic evidence around here lately seems to not matter and is ignored too when posted by certain individuals eh.